# Working with a Tektronix TDS 3012B Oscilloscope EE 310: ELECTRONIC CIRCUIT DESIGN I

Size: px
Start display at page:

Download "Working with a Tektronix TDS 3012B Oscilloscope EE 310: ELECTRONIC CIRCUIT DESIGN I"

## Transcription

1 Working with a Tektronix TDS 3012B Oscilloscope EE 310: ELECTRONIC CIRCUIT DESIGN I Prepared by: Kyle Botteon Questions? 2

2 Background Information Recall that oscilloscopes (scopes) are specialized measurement tools. While a digital multi-meter can collect measurements that are constant with time, an oscilloscope can display and measure a time-varying signal. Moreover, it can capture and present more than one signal simultaneously for comparison or to apply mathematical operations. Keep a few things in mind while working: 1. Scopes have protection circuitry, but everything has its limits. Read warnings before performing the steps that follow to avoid breaking equipment. 2. Measurement tools, no matter how well-designed, will change the circuit once one is introduced. So, measurements will also be influenced to some degree. Always consider the effects of adding a measurement tool to a circuit. For the exercises in this guide, the oscilloscope is set to behave like a 1MΩ resistor between the node under observation and circuit ground. Materials Tektronix TDS 3012B Oscilloscope A suitable circuit to examine: o 50mV < Peak Voltage < 50V o Frequency < 25 MHz o Wired on a breadboard BNC-to-alligator or BNC-to-clip/grabber probe(s); the scope will support up to 2 measurement probes BNC-to-alligator Probe BNC Connector Grabber Connector Alligator Connector 3

4 Capturing a Signal With some luck, a signal might just show up once a probe is connected to the circuit. But, odds are, the timescale, vertical scale, and other settings are not configured properly. For this exercise, a 4.5V PP, 250 KHz, +1.5V DC triangle wave is applied to the CH 1 input, set to 1MΩ mode. Scaling of Offsetting Notice the timescale in the bottom center of the display (the number with units in time). If it is not % of the period of the signal under observation, something like the first image to the right might appear. 1. Adjust the timescale to be a reasonable 1 number using the SCALE knob in the HORIZONTAL settings group. Ensure the signal to be adjusted (CH 1 or CH 2 probe) is selected by pressing the CH 1 or CH 2 button before changing the scale. After properly adjusting the horizontal scale, the entire signal may not be onscreen we must adjust the vertical scale too. 2. Adjust the vertical scale to be a reasonable 2 number using the SCALE knob in the VERTICAL settings group. Ensure the signal to be adjusted (CH 1 or CH 2 probe) is selected by pressing the CH 1 or CH 2 button before changing the scale. 3. Use the POSITION knobs in the HORIZONTAL and VERTICAL setting groups to adjust the position of the wave on the display. For example, the waveform could be centered on the screen. 4.5VPP, 250 KHz, +1.5V DC triangle wave with default settings 4.5VPP, 250 KHz, +1.5V DC triangle wave with appropriate timescale 4.5VPP, 250 KHz, +1.5V DC triangle wave with appropriate vertical scale 1 Circuit-dependent. Base a starting point on the expected signal period. For instance, for a 100 Hz signal (with period 10 ms), a timescale of 2 10ms is appropriate. 2 Circuit-dependent. Use an estimated figure as a starting point then further adjust the scale to suit. 5

5 Adjusting the Trigger The trigger setting controls when the scope will capture the waveform applied to a probe. If the trigger is larger than the signal peaks, the scope will never capture a waveform. If it is too low, the scope will trigger on noise, capturing many waveforms and presenting a distorted image onscreen. For this exercise, a 2.5V PP, 250 KHz, +0V DC square wave is applied to the CH 1 input, set to 1MΩ mode. The default trigger setting, trigger on rising edge, is assumed. 1. Observe the right hand side of the gridline-filled portion of the display. A yellow arrow, pointing left, indicates the trigger level, according to the vertical scale of the display. 2. Twist the LEVEL knob in the TRIGGER setting group until the yellow arrow points to the amplitude at which the signal should be captured. For periodic signals, it is wise to set the trigger to about 75% of the maximum amplitude of the waveform. By doing so, undesirable effects of noise are mitigated. For other signals, a guess-and-check method is suitable. Poorly adjusted trigger, set to about 1V The AUTOSET Button There exists a button that can configure all of the previous setting automatically at a price. The AUTOSET button runs a software routine that locates a signal, scales and offsets it, and adjusts the trigger. But, high frequency noise and other factors can cause the incorrect settings to be applied. For this reason, blindly accepting the result is a mistake. Still, it is a useful tool when used properly. 1. Verify at least one probe is connected to a circuit node under observation. 2. Press the CH 1 button if automatic settings should be applied to channel 1, otherwise press the CH 2 button. 3. Press the AUTOSET button. At this point, it is highly likely that a properlyadjusted signal is displayed onscreen. Still, verify the settings that were automatically applied: 4. Check the timescale and compare it to the expected timescale for the signal under observation. For example, if a 100 Hz signal (with period 10 ms) is applied to CH 1, and CH 1 is selected when AUTOSET is pressed, a timescale in the range 1 10 ms should have been configured. 6 Well-adjusted trigger, set to about 750mV Poor AUTOSET result (applied to a 10Hz triangle wave with 50% output noise) The above example demonstrates why AUTOSET is not always reliable. The 10 Hz (period 100ms) signal under observation is buried in noise, so the automatic settings adjusted the timescale to a short (10µS) period.

6 Measuring a Signal The scope has software-based measurement tools available to observe and quantify signal parameters. These are especially useful for collecting exact figures, as opposed to estimating them with the gridlines onscreen. Automatic Measurements To apply an automatic measurement: 1. Select CH 1 or CH 2 based which channel a measurement should be applied to. 2. Press the MEASURE button. A measurements menu will appear. 3. Use the -MORE- button to cycle through the six menus of measurements. 4. Choose a measurement, then apply it by pressing the button next to it. A measurement name and numerical value will appear in the color corresponding to the channel it is applied to (e.g. yellow for CH 1) 5. Press the MENU OFF button to close the measurements menu and move measurements to the side of the display. Manual Measurements Automatic measurements can be erroneous, especially when applied to noisy signals. For this reason, manual measurement tools are available. H BARS and V BARS place horizontal and vertical lines onscreen and measure the difference between the values they represent. To enable these tools: 1. Press the CURSOR button on the top right of the scope. A cursor function menu will appear. 2. Press the button associated with H BARS or V BARS based on what will be measured. 3. Adjust the location of the bars using the SELECT, COURSE, and unlabeled knob on the top center of the scope. SELECT is used to switch between the two bars, while COURSE changes the speed at which the bars may be moved with the unlabeled knob. Automatic measurement menu Automatic measurements applied The position of each bar and the difference between them is located in the top right of the display. H Bars and V Bars tools 7

7 Troubleshooting Sometimes things don t always go as planned. Consider the following tips when the scope doesn t behave as expected. General Tips Verify factory defaults were recalled prior to configuring the scope or taking any measurements. Almost always, the previous user s settings are still programmed. Check the circuit configuration. Even a single missing wire or incorrect connection could change results entirely. If no signal appears onscreen, even after scaling and offsetting Check for loose connections between the probe alligator clips or grabber clips. Verify the black leads of the probes are connected to circuit ground. If earth ground is introduced, verify the black leads of the probes and earth ground are at the same potential. If an unexpected signal appears onscreen Check the timescale a small piece of a larger waveform might be displayed. Verify the probe clips are connected to the correct circuit node. If all else fails Recall factory defaults once more and try again. Call a TA over and explain the problem. 8

### Oscilloscope Guide Tektronix TDS3034B & TDS3052B

Tektronix TDS3034B & TDS3052B Version 2008-Jan-1 Dept. of Electrical & Computer Engineering Portland State University Copyright 2008 Portland State University 1 Basic Information This guide provides basic

### Quick Start. RSHS1000 Series Handheld Digital Oscilloscope

Quick Start RSHS1000 Series Handheld Digital Oscilloscope General Safety Summary Carefully read the following safety precautions to avoid personal injury and prevent damage to the instrument or any products

### ME EN 363 ELEMENTARY INSTRUMENTATION Lab: Basic Lab Instruments and Data Acquisition

ME EN 363 ELEMENTARY INSTRUMENTATION Lab: Basic Lab Instruments and Data Acquisition INTRODUCTION Many sensors produce continuous voltage signals. In this lab, you will learn about some common methods

### BME 3512 Biomedical Laboratory Equipment List

BME 3512 Biomedical Laboratory Equipment List Agilent E3630A DC Power Supply Agilent 54622A Digital Oscilloscope Agilent 33120A Function / Waveform Generator APPA 95 Digital Multimeter Component Layout

### University of Utah Electrical & Computer Engineering Department ECE1050/1060 Oscilloscope

University of Utah Electrical & Computer Engineering Department ECE1050/1060 Oscilloscope Name:, A. Stolp, 2/2/00 rev, 9/15/03 NOTE: This is a fill-in-the-blanks lab. No notebook is required. You are encouraged

### Technical Reference. TDS 684A, TDS 744A, & TDS 784A Digitizing Oscilloscope Performance Verification and Specifications

Technical Reference TDS 684A, TDS 744A, & TDS 784A Digitizing Oscilloscope Performance Verification and Specifications 070-8990-04 Please check for change information at the rear of this manual. Fifth

### Experiment # 4 Counters and Logic Analyzer

EE20L - Introduction to Digital Circuits Experiment # 4. Synopsis: Experiment # 4 Counters and Logic Analyzer In this lab we will build an up-counter and a down-counter using 74LS76A - Flip Flops. The

### Advanced Skills with Oscilloscopes

Advanced Skills with Oscilloscopes A Hands On Laboratory Guide to Oscilloscopes using the Rigol DS1104Z By: Tom Briggs, Department of Computer Science & Engineering Shippensburg University of Pennsylvania

### Burlington County College INSTRUCTION GUIDE. for the. Hewlett Packard. FUNCTION GENERATOR Model #33120A. and. Tektronix

v1.2 Burlington County College INSTRUCTION GUIDE for the Hewlett Packard FUNCTION GENERATOR Model #33120A and Tektronix OSCILLOSCOPE Model #MSO2004B Summer 2014 Pg. 2 Scope-Gen Handout_pgs1-8_v1.2_SU14.doc

### TDS 520B, TDS 540B, TDS 620B, TDS 644B, TDS 680B, TDS 684B, TDS 724A, TDS 744A, & TDS 784A

Technical Reference TDS 520B, TDS 540B, TDS 620B, TDS 644B, TDS 680B, TDS 684B, TDS 724A, TDS 744A, & TDS 784A Digitizing Oscilloscopes Performance Verification and Specifications 070-9384-01 Copyright

### INSTRUCTION GUIDE. for the. Keysight (formally Agilent; Hewlett-Packard) Model #33509B. and. Tektronix. Fall 2017

v3 INSTRUCTION GUIDE for the Keysight (formally Agilent; Hewlett-Packard) WAVEFORM GENERATOR Model #33509B 33500 Series and Tektronix OSCILLOSCOPE Model #MSO2004B Fall 2017 Pg. 2 Scope-Gen Handout_part

### Spectrum Analyser Basics

Hands-On Learning Spectrum Analyser Basics Peter D. Hiscocks Syscomp Electronic Design Limited Email: phiscock@ee.ryerson.ca June 28, 2014 Introduction Figure 1: GUI Startup Screen In a previous exercise,

### Viewing Serial Data on the Keysight Oscilloscopes

Ming Hsieh Department of Electrical Engineering EE 109L - Introduction to Embedded Systems Viewing Serial Data on the Keysight Oscilloscopes by Allan G. Weber 1 Introduction The four-channel Keysight (ex-agilent)

### Beginners How to Test DSO138mini

Beginners How to Test DSO138mini You have finished assembling your DSO138mini kit. You may be anxious to see it works. But you might not be familiar with oscilloscope and you could encounter unexpected

### Agilent DSO5014A Oscilloscope Tutorial

Contents UNIVERSITY OF CALIFORNIA AT BERKELEY College of Engineering Department of Electrical Engineering and Computer Sciences EE105 Lab Experiments Agilent DSO5014A Oscilloscope Tutorial 1 Introduction

### Viewing Serial Data on the Keysight Oscilloscopes

Ming Hsieh Department of Electrical Engineering EE 109L - Introduction to Embedded Systems Viewing Serial Data on the Keysight Oscilloscopes by Allan G. Weber 1 Introduction The four-channel Keysight (ex-agilent)

Hello, welcome to Analog Arts spectrum analyzer tutorial. Please feel free to download the Demo application software from analogarts.com to help you follow this seminar. For this presentation, we use a

### MINI PC SCOPE PCSU01. User manual. test leads software download USB cable design enclosure

MINI PC SCOPE PCSU01 User manual Features test leads software download USB cable design enclosure Specifications oscilloscope: o bandwidth: DC to 200 khz ± 3 db o input impedance: 100 ko / 20 pf o maximum

### University of Utah Electrical Engineering Department EE1050/1060 Oscilloscope. Name:, Lab TA:

University of Utah Electrical Engineering Department EE1050/1060 Oscilloscope Name:, Lab TA: A. Stolp, 2/2/00 rev, 9/14/00 NOTE: This is a fill-in-the-blanks lab. No notebook is required. You are encouraged

### PB-507. Advanced Analog & Digital Electronic Design Workstation Instruction Manual. Revision: 2/2014

PB-507 Advanced Analog & Digital Electronic Design Workstation Instruction Manual Revision: 2/2014 Test Equipment Depot - 800.517.8431-99 Washington Street Melrose, MA 02176 TestEquipmentDepot.com 1 1

### Electrical and Electronic Laboratory Faculty of Engineering Chulalongkorn University. Cathode-Ray Oscilloscope (CRO)

2141274 Electrical and Electronic Laboratory Faculty of Engineering Chulalongkorn University Cathode-Ray Oscilloscope (CRO) Objectives You will be able to use an oscilloscope to measure voltage, frequency

### S op o e p C on o t n rol o s L arni n n i g n g O bj b e j ctiv i e v s

ET 150 Scope Controls Learning Objectives In this lesson you will: learn the location and function of oscilloscope controls. see block diagrams of analog and digital oscilloscopes. see how different input

### imso-104 Manual Revised August 5, 2011

imso-104 Manual Revised August 5, 2011 Section 1 Getting Started SAFETY 1.10 Quickstart Guide 1.20 SAFETY 1.30 Compatibility 1.31 Hardware 1.32 Software Section 2 How it works 2.10 Menus 2.20 Analog Channel

Page 1/9 Waveforms 2015 tutorial 3-Jan-18 Table of Contents Introduction Introduction to DAD/NAD and Waveforms 2015... 2 Digital Functions Static I/O... 2 LEDs... 2 Buttons... 2 Switches... 2 Pattern Generator...

### Lab 2: A/D, D/A, and Sampling Theorem

Lab 2: A/D, D/A, and Sampling Theorem Introduction The purpose of this lab is to explore the principles of analog-to-digital conversion, digital-to-analog conversion, and the sampling theorem. It will

### Troubleshooting Your Design with the TDS3000C Series Oscilloscopes

Troubleshooting Your Design with the 2 Table of Contents Getting Started........................................................... 4 Debug Digital Timing Problems...............................................

### Physics 123 Hints and Tips

Physics 123 Hints and Tips Solderless Breadboards All of the analog labs and most of the digital labs will be built on the Proto-Board solderless breadboards. These provide three solderless breadboard

### Introduction to the oscilloscope and digital data acquisition

Introduction to the oscilloscope and digital data acquisition Eric D. Black California Institute of Technology v1.1 There are a certain number of essential tools that are so widely used that every aspiring

### imso-104 Manual Revised July 19, 2012

imso-104 Manual Section 1 Getting Started SAFETY 1.10 Quickstart Guide 1.20 SAFETY 1.30 Compatibility 1.31 Hardware 1.32 Software Section 2 How it works 2.10 Menus 2.20 Analog Channel 2.21 On / Off 2.22

### User Manual. Digital Storage Oscilloscopes Models 2534, 2540 & 2542

User Manual Digital Storage Oscilloscopes Models 2534, 2540 & 2542 General Safety Summary General Safety Summary Review the following safety precautions to avoid injury and prevent damage to this product

### Manual Supplement. This supplement contains information necessary to ensure the accuracy of the above manual.

Manual Title: 9500B Users Supplement Issue: 2 Part Number: 1625019 Issue Date: 9/06 Print Date: October 2005 Page Count: 6 Version 11 This supplement contains information necessary to ensure the accuracy

### Overview. Know Your Oscilloscope. Front Panel. Rear Panel. Sharing Agilent s Resources with Engineering Educators

Know Your Oscilloscope Overview Front Panel Sharing Agilent s Resources with Engineering Educators www.educatorscorner.com Horizontal (time) controls Run control Special purpose menus/controls Trigger

### User Manual. Digital Storage Oscilloscopes Models 2534, 2540 & General Safety Summary. Version 1.03

General Safety Summary General Safety Summary User Manual Digital Storage Oscilloscopes Models 2534, 2540 & 2542 Review the following safety precautions to avoid injury and prevent damage to this product

### MSO-28 Oscilloscope, Logic Analyzer, Spectrum Analyzer

Link Instruments Innovative Test & Measurement solutions since 1986 Store Support Oscilloscopes Logic Analyzers Pattern Generators Accessories MSO-28 Oscilloscope, Logic Analyzer, Spectrum Analyzer \$ The

### Measuring with digital storage oscilloscopes

Markus Reil, Rainer Wagner 2.2016 1MAA265-1e Educational note Measuring with digital storage oscilloscopes Educational note Products: ı R&S HMO1002 This educational note covers the theory and practice

### Quick Start. SHS1000 Series Handheld Digital Oscilloscope QS03010-E02B 2015 SIGLENT TECHNOLOGIES CO., LTD

Quick Start SHS1000 Series Handheld Digital Oscilloscope QS03010-E02B 2015 SIGLENT TECHNOLOGIES CO., LTD Guaranty and Declaration Copyright SIGLENT TECHNOLOGIES CO., LTD. All Rights Reserved. Trademark

### Digital Delay / Pulse Generator DG535 Digital delay and pulse generator (4-channel)

Digital Delay / Pulse Generator Digital delay and pulse generator (4-channel) Digital Delay/Pulse Generator Four independent delay channels Two fully defined pulse channels 5 ps delay resolution 50 ps

### ELECTRICAL ENGINEERING DEPARTMENT California Polytechnic State University

EECTRICA ENGINEERING DEPARTMENT California Polytechnic State University EE 361 NAND ogic Gate, RS Flip-Flop & JK Flip-Flop Pre-lab 7 1. Draw the logic symbol and construct the truth table for a NAND gate.

### Exercise 2: D-Type Flip-Flop

Flip-Flops Digital Logic Fundamentals Exercise 2: D-Type Flip-Flop EXERCISE OBJECTIVE When you have completed this exercise, you will be able to determine the characteristics of a D-type results with an

### PRELIMINARY INFORMATION. Professional Signal Generation and Monitoring Options for RIFEforLIFE Research Equipment

Integrated Component Options Professional Signal Generation and Monitoring Options for RIFEforLIFE Research Equipment PRELIMINARY INFORMATION SquareGENpro is the latest and most versatile of the frequency

### GDB-03 Demo Module USER MANUAL GW INSTEK PART NO. 82DB-03000M01 ISO-9001 CERTIFIED MANUFACTURER

GDB-03 Demo Module USER MANUAL GW INSTEK PART NO. 82DB-03000M01 ISO-9001 CERTIFIED MANUFACTURER This manual contains proprietary information, which is protected by copyright. All rights are reserved. No

### WAVEJET 300 SERIES OSCILLOSCOPES. New Cover to Come. Unmatched Performance, Portability, and Value

WAVEJET 300 SERIES OSCILLOSCOPES New Cover to Come Unmatched Performance, Portability, and Value ALL THE TOOLS YOU NEED Automatic Measurements Save time making measurements on your signals by using the

### Troubleshooting Analog to Digital Converter Offset using a Mixed Signal Oscilloscope APPLICATION NOTE

Troubleshooting Analog to Digital Converter Offset using a Mixed Signal Oscilloscope Introduction In a traditional acquisition system, an analog signal input goes through some form of signal conditioning

### MIE 402: WORKSHOP ON DATA ACQUISITION AND SIGNAL PROCESSING Spring 2003

MIE 402: WORKSHOP ON DATA ACQUISITION AND SIGNAL PROCESSING Spring 2003 OBJECTIVE To become familiar with state-of-the-art digital data acquisition hardware and software. To explore common data acquisition

### Fluke /

Fluke 190-104/190-204 ScopeMeter 190 Series II Service Manual PN 4822 872 05405 March 2011 2011 Fluke Corporation, All rights reserved. Printed in the Netherlands All product names are trademarks of their

### Analog Discovery Scope and Waveform Generator Edited 11/15/2016 by Eric Scotti & DGH

Analog Discovery Scope and Waveform Generator Edited 11/15/2016 by Eric Scotti & DGH Specifications The Analog Discovery contains several devices but we will likely only use the 2 channel oscilloscope

### WAVEJET 300 SERIES OSCILLOSCOPES. Unmatched Performance, Portability, and Value

WAVEJET 300 SERIES OSCILLOSCOPES Unmatched Performance, Portability, and Value 1 WAVEJET 300 SERIES Unique Capabilities in a Low Bandwidth Oscilloscope The WaveJet 300 Series features unmatched performance

### Digitizing Oscilloscopes

Digitizing Oscilloscopes This document is meant to be a reference to the operation of the digitizing oscilloscopes available in the laboratories. Major topics will be covered, but not all the features

### Basic LabVIEW Programming Amit J Nimunkar, Sara Karle, Michele Lorenz, Emily Maslonkowski

Introduction This lab familiarizes you with the software package LabVIEW from National Instruments for data acquisition and virtual instrumentation. The lab also introduces you to resistors, capacitors,

### System Troubleshooting for

Brought to You by Presented by Part 2 of 4 A1 Part 2 of 4 Video Problems No Match for the TERMINATOR When electronic security systems fail or malfunction, more often than not it s something small and relatively

### Contents: 1 LANsmart Pro Main Unit 4 Remote Unit: ID1, ID2, ID3, ID4

LANsmart Pro user manual Introduction LANsmart Pro is a hand-held, multifunction Cable Map Tester and Cable Length Meter. It has an integrated Analog and Digital Tone Generator, Port Finder, and Quick

### FREQUENCY COUNTERS TO 18 GHZ USING THE DATUM FREQUENCY STANDARD

TECHNICAL MANUAL AF-166 INSTRUMENT CALIBRATION FREQUENCY COUNTERS TO 18 GHZ USING THE DATUM 9390-6000-34 FREQUENCY STANDARD THIS PUBLICATION SUPERSEDES NAVAIR 17-20AF-166 DATED 1 FEBRUARY 2005 DISTRIBUTION

### Serial Decode I2C TEN MINUTE TUTORIAL. December 21, 2011

Serial Decode I2C TEN MINUTE TUTORIAL December 21, 2011 Summary LeCroy oscilloscopes have the ability to trigger on and decode multiple serial data protocols. The decode in binary, hex, or ASCII format,

### PicoScope 6407 Digitizer

YE AR PicoScope 6407 Digitizer HIGH PERFORMANCE USB DIGITIZER Programmable and Powerful 1 GHz bandwidth 1 GS buffer size 5 GS/s real-time sampling Advanced digital triggers Built-in function generator

### 2510 Series Handheld Digital Storage Oscilloscopes

Model: 2511, 2512, 2515, 2516 2510 Series Handheld Digital Storage Oscilloscopes USER MANUAL Safety Summary The following safety precautions apply to both operating and maintenance personnel and must be

### "shell" digital storage oscilloscope (Beta)

"shell" digital storage oscilloscope (Beta) 1. Main board: solder the element as the picture shows: 2. 1) Check the main board is normal or not Supply 9V power supply through the connector J7 (Note: The

### Fluke 190-Series II Firmware Upgrade V11.44

Fluke 190-Series II Firmware Upgrade V11.44 Requirements 1. Fluke 190- Series II ScopeMeter with firmware prior to V11.44 2. Supported models are: 190-102, 190-104, 190-062, 190-202, 190-204, 190-502,

### INTRODUCTION This procedure should only be performed if the instrument fails to meet the Performance Check tests for Output Zero or Offset Accuracy

INTRODUCTION This procedure should only be performed if the instrument fails to meet the Performance Check tests for Output Zero or Offset Accuracy (steps A and B). Gain, which affects DC Accuracy, cannot

### RICHLAND COLLEGE School of Engineering Business & Technology Rev. 0 W. Slonecker Rev. 1 (8/26/2012) J. Bradbury

RICHLAND COLLEGE School of Engineering Business & Technology Rev. 0 W. Slonecker Rev. 1 (8/26/2012) J. Bradbury INTC 1307 Instrumentation Test Equipment Teaching Unit 8 Oscilloscopes Unit 8: Oscilloscopes

### Digital Storage Oscilloscopes 2550 Series

Data Sheet Digital Storage Oscilloscopes 2550 Series The 2550 series digital storage oscilloscopes provide high performance and value in 2-channel and 4-channel configurations. With bandwidth from 70 MHz

### RIGOL. Data Sheet. DS1000B Series Digital Oscilloscopes DS1074B, DS1104B, DS1204B. Product Overview. Easy to Use Design. Applications.

RIGOL Data Sheet Product Overview DS1000B series oscilloscopes are designed with four analog channels and 1 external trigger channel, which can capture multi-channel signal simultaneously and meet industrial

### Troubleshooting Your Design with Tektronix MSO and DPO Series Oscilloscopes

Troubleshooting Your Design with Tektronix MSO and DPO Series Oscilloscopes Our thanks to Tektronix for allowing us to reprint the following article. Today s engineers and technicians face increasingly

### Exercise 5-1. Troubleshooting Techniques EXERCISE OBJECTIVE DISCUSSION OUTLINE DISCUSSION. Signal flow tracing

Exercise 5-1 Troubleshooting Techniques EXERCISE OBJECTIVE When you have completed this exercise, you will be able to apply a systematic technique of signal flow tracing to diagnose instructor-inserted

### Analog Arts SA985 SA975 SA935 SA915 Product Specifications [1]

www.analogarts.com Analog Arts SA985 SA975 SA935 SA915 Product Specifications [1] 1. These models consist of an oscilloscope, a spectrum analyzer, a data recorder, and a frequency & phase meter. Oscilloscope/

### Expect to Make Waves.

Expect to Make Waves. The New Oscilloscope Large 10.4" LCD touch screen Long capture time Extensive communication capabilities www.lecroy.com The New Oscillos From its large 10.4" LCD touch screen to its

### Instructions. P MHz 1X/10X Passive Probe

Instructions P2100 100 MHz 1X/10X Passive Probe 071-0774-01 071077401 Copyright Tektronix, Inc. All rights reserved. Tektronix products are covered by U.S. and foreign patents, issued and pending. Information

### MAIN PCA TEST PROCEDURE DOCUMENT NUMBER:

TEST PROCEDURE DOCUMENT NUMBER: -20 Signature Date Author J. Hoo 10-7-96 CHK J. Hoo 10-7-96 CHK - - Approved J. Hoo 10-7-96 Huntron Instruments, Inc. 1996 All rights reserved. This document may not be

### Copyright 2018 Geophysical Survey Systems, Inc. All rights reserved including the right of reproduction in whole or in part in any form

MN36-268 Rev A Copyright 2018 Geophysical Survey Systems, Inc. All rights reserved including the right of reproduction in whole or in part in any form Published by Geophysical Survey Systems, Inc. 40 Simon

### PicoScope 6407 Digitizer

YE AR HIGH PERFORMANCE USB DIGITIZER Programmable and Powerful 1 GHz bandwidth 1 GS buffer size 5 GS/s real-time sampling Advanced digital triggers Built-in function generator USB-connected Signals Analysis

### Using an oscilloscope - The Hameg 203-6

Using an oscilloscope - The Hameg 203-6 What does an oscilloscope do? Setting up How does an oscilloscope work? Other oscilloscope controls Connecting a function generator Microphones audio signals and

### User Manual. TDS3VID Extended Video Application Module

User Manual TDS3VID Extended Video Application Module 071-0328-02 071032802 Copyright Tektronix. All rights reserved. Licensed software products are owned by Tektronix or its subsidiaries or suppliers,

### Application Note AN-708 Vibration Measurements with the Vibration Synchronization Module

Application Note AN-708 Vibration Measurements with the Vibration Synchronization Module Introduction The vibration module allows complete analysis of cyclical events using low-speed cameras. This is accomplished

### Experiment 9A: Magnetism/The Oscilloscope

Experiment 9A: Magnetism/The Oscilloscope (This lab s "write up" is integrated into the answer sheet. You don't need to attach a separate one.) Part I: Magnetism and Coils A. Obtain a neodymium magnet

### Fluke 19xB-19xC-2x5C. Service Manual. ScopeMeter. models 192B,196B,199B,192C,196C,199C,215C,225C

Fluke 19xB-19xC-2x5C ScopeMeter models 192B,196B,199B,192C,196C,199C,215C,225C Service Manual PN 4822 872 05391 September 2002, rev.2, Oct.09 2002-2009 Fluke Corporation, All rights reserved. All product

### Reference. TDS7000 Series Digital Phosphor Oscilloscopes

Reference TDS7000 Series Digital Phosphor Oscilloscopes 07-070-00 0707000 To Use the Front Panel You can use the dedicated, front-panel knobs and buttons to do the most common operations. Turn INTENSITY

### R&S RT-ZF20 Power Deskew Fixture User Manual

R&S RT-ZF20 Power Deskew Fixture User Manual (B00X2) User Manual Test & Measurement 1800.0040.02 04 This manual describes the following R&S RT-ZF models: R&S RT-ZF20 (1800.0004.01) 2016 Rohde & Schwarz

### Cryoelectronics. MS-FLL User s Manual. Mr. SQUID Flux-Locked Loop. STAR Cryoelectronics 25 Bisbee Court, Suite A Santa Fe, NM U. S. A.

Cryoelectronics MS-FLL User s Manual Mr. SQUID Flux-Locked Loop STAR Cryoelectronics 25 Bisbee Court, Suite A Santa Fe, NM 87508 U. S. A. STAR Cryoelectronics, LLC ii Table of Contents Revision Record...

### 4830A Accelerometer simulator Instruction manual. IM4830A, Revision E1

4830A Accelerometer simulator Instruction manual IM4830A, Revision E1 IM4830, Page 2 The ENDEVCO Model 4830A is a battery operated instrument that is used to electronically simulate a variety of outputs

### User s Manual. TDO1000/TDO2000 Series Oscilloscopes

User s Manual TDO1000/TDO2000 Series Oscilloscopes Manual Print History The manual print history shown below lists all the printing dates and editions. The printing date changes when a new edition is released.

### User Manual. Digital Oscilloscope V1.9

User Manual Digital Oscilloscope V1.9 Declaration All rights reserved. Contents in this Manual are not allowed to copy, extract and translate before being allowed by the company. Brief Introduction Characteristic:

### Model P2016. Dual Channel Handheld Digital Storage. Oscilloscope & Multimeter. Quick Scope Guide

Model P2016 Dual Channel Handheld Digital Storage Oscilloscope & Multimeter Quick Scope Guide General Warranty BNC warrants that the product will be free from defects in materials and workmanship for 3

### Metal Electrode Meter

Metal Electrode Meter INSTRUCTION MANUAL FOR Metal Electrode Meter MODEL 2900 Serial # Date PO Box 850 Carlsborg, WA 98324 U.S.A. 360-683-8300 800-426-1306 FAX: 360-683-3525 http://www.a-msystems.com Version

### Analog Arts SA985 SA975 SA935 SA915 Product Specifications

Analog Arts SA985 SA975 SA935 SA915 Product Specifications Oscilloscope/ Spectrum Analyzer/ Data Recorder Model SA985 SA975 SA935 SA915 Oscilloscope (Typical by Design) Bandwidth (Max at probe tip) [1]

### Dave Jones Design Phone: (607) Lake St., Owego, NY USA

Manual v1.00a June 1, 2016 for firmware vers. 2.00 Dave Jones Design Phone: (607) 687-5740 34 Lake St., Owego, NY 13827 USA www.jonesvideo.com O Tool Plus - User Manual Main mode NOTE: New modules are

### The Measurement Tools and What They Do

2 The Measurement Tools The Measurement Tools and What They Do JITTERWIZARD The JitterWizard is a unique capability of the JitterPro package that performs the requisite scope setup chores while simplifying

### APPLICATION NOTE # Monitoring DTMF Digits Transmitted by a Phone

APPLICATION NOTE # Product: 930A Communications Test Set 930i Communications Test Set Monitoring DTMF Digits Transmitted by a Phone Introduction This Application Note describes how to configure and connect

### 25 MHz Digital Storage Oscilloscope

99 Washington Street Melrose, MA 02176 Phone 781-665-1400 Toll Free 1-800-517-8431 Visit us at www.testequipmentdepot.com Model: 2530B 25 MHz Digital Storage Oscilloscope USER MANUAL Safety Summary The

### Getting Started with the LabVIEW Sound and Vibration Toolkit

1 Getting Started with the LabVIEW Sound and Vibration Toolkit This tutorial is designed to introduce you to some of the sound and vibration analysis capabilities in the industry-leading software tool

### Experiment 9 Analog/Digital Conversion

Experiment 9 Analog/Digital Conversion Introduction Most digital signal processing systems are interfaced to the analog world through analogto-digital converters (A/D) and digital-to-analog converters

### Activity P32: Variation of Light Intensity (Light Sensor)

Activity P32: Variation of Light Intensity (Light Sensor) Concept DataStudio ScienceWorkshop (Mac) ScienceWorkshop (Win) Illuminance P32 Vary Light.DS P54 Light Bulb Intensity P54_BULB.SWS Equipment Needed

### 4CH/2CH Digital Storage Oscilloscopes

Model: 2553, 2555, 2556, 2557, 2558, 2559 4CH/2CH Digital Storage Oscilloscopes USER MANUAL Safety Summary The following safety precautions apply to both operating and maintenance personnel and must be

### LabView Exercises: Part II

Physics 3100 Electronics, Fall 2008, Digital Circuits 1 LabView Exercises: Part II The working VIs should be handed in to the TA at the end of the lab. Using LabView for Calculations and Simulations LabView

### User Manual. TDS3SDI 601 Digital Video Application Module

User Manual TDS3SDI 601 Digital Video Application Module 071-0787-00 071078700 Copyright Tektronix, Inc. All rights reserved. Tektronix products are covered by U.S. and foreign patents, issued and pending.

### USER MANUAL. Blackburst, Sync, Audio Tone Generator. For Models BSG-50, RM-50/BSG, SR-50/BSG. Doc Rev. F (C) Copyright 2014

HORITA BSG-50 Blackburst, Sync, Audio Tone Generator USER MANUAL For Models BSG-50, RM-50/BSG, SR-50/BSG Doc. 070450 Rev. F (C) Copyright 2014 P.O. Box 3993, Mission Viejo, CA 92690 (949) 489-0240 www.horita.com

### Lab experience 1: Introduction to LabView

Lab experience 1: Introduction to LabView LabView is software for the real-time acquisition, processing and visualization of measured data. A LabView program is called a Virtual Instrument (VI) because

### Agilent N6467A BroadR-Reach Compliance Test Application. Methods of Implementation

Agilent N6467A BroadR-Reach Compliance Test Application Methods of Implementation s1 Notices Agilent Technologies, Inc. 2013 No part of this manual may be reproduced in any form or by any means (including

### Operating Instructions

Operating Instructions HAEFELY TEST AG KIT Measurement Software Version 1.0 KIT / En Date Version Responsable Changes / Reasons February 2015 1.0 Initial version WARNING Introduction i Before operating

### User Manual. GPS-1000B+ series. Digital Oscilloscope V 1.1

User Manual GPS-1000B+ series Digital Oscilloscope V 1.1 Brief Introduction Characteristic: The volume of the oscilloscope is cabinet and it is portable 7 Color TFT LCD display 2 channels, Bandwidth: